1. The flow direction is the same as our factory arrow.
2. Electromagnetic flowmeter can be installed horizontally or vertically. If it is installed vertically, it should be flowed from the bottom to the top. Inclined pipe, from top to bottom. If it is from top to bottom, the exhaust valve is generally installed. Generally, it is not installed on the pipeline. Place, avoid empty tubes or air bubbles.
3. Can not be installed on the suction side of the pump, that is, it cannot be installed in the pipeline with negative pressure. Lining will bulge.