M66 X 3
Split type high quality and reliability ultrasonic water level sensor introduction
Ultrasonic water level sensor is a non-contact liquid level gauge. It can be used to measure liquid level and flow in various containers or pipelines. It can also be used in the measurement of water level in drains, reservoirs, rivers and lakes. It is especially suitable for sewage and corrosive applications such as urban drainage pumps. Measurement of the water level before and after the trash rack.
Split type high quality and reliability ultrasonic water level sensor detail
Original: Beijing, China
Certification: ISO9001,CE etc
Voltage: 24VDC/ 220VAC
Payment term: 30% T/T in advance,70%T/T before delivery date
Delivery date: 3~7days, according to your order
Packing: wooden box+paper box
Market: Middle East/ Africa/ Asia/ South America/ Europe/ North America
Warranty: 1 year
WE SUPPORT PERSONALIZED DESIGN AND OEM BASIS COOPERATION
Split type high quality and reliability ultrasonic water level sensor characteristics
1. Ultrasonic transmission speed is relatively stable
2. No contact with the medium, no moving parts, the electronic components only vibrate with audible frequencies, small amplitude, long instrument life;
3. Not only continuous measurement and fixed-point measurement, but also easy to provide remote measurement or remote-control signal;
4. Can measure the liquid level of high-speed or tilted liquid
Split type high quality and reliability ultrasonic water level sensor trouble shooting
Symptoms Causes of failure Solution
1. Instrument does not display, does not work Power supply error wiring error Check DC24v power supply is correct Check wiring is correct
The instrument has a display, does not work level gauge is not aligned liquid level or material surface fluctuations is very large material surface is not smooth surface has a thick layer of foam liquid emptying, the bottom of the container after the material is empty is not flat out of measurement The range is adjusted by aligning the level meter, and the horizontal calibration can be used to add the plastic pipe in the container (see the installation section). Use a more large-scale level meter to add more liquid or other measurement methods. Naturally restores work after feeding and switches to a more extensive level gauge
2. Instrument display is unstable or there is a large deviation in the measured value. Entry into the blind zone. Unstable power supply. Strong electromagnetic interference. Suppression of sound waves. Preparation of metal flanges on the level gauge. Probe surface or side contact with metal. Load level gauge or prevent over-prevention of material level. Improve the power supply to the level gauge. Connect to earth or shield to change the installation position or add plastic pipe. Use plastic flange instead. Use rubber pad to isolate metal.
1. Are you a manufacturer, trading company or a third party?
A: We are a manufacturer, and we have built up our company since 2005
2. Where is your factory located?
A: Our factory address is: Room410 TowerC Jiahua Buiding NO.9 Shangdi 3 street Haidian district Beijing
3.How can I get the product information?
A: Beijing Gallop Technology co.,ltd
Whatsapp no: 0086 13858586115
4. What is your terms of packing?
A: Generally, we pack our goods in wooden box and paper box.
5. What is your terms of payment?
A: For large order T/T 30% in advance, and 70% before delivery. For sample and small order in 100% payment.
6. What is your terms of delivery?
A: FOB, CNF, CIF and so on.
7. How can i get the after-service?
A: We will send you the spare parts by free if the problems caused by us. If it is the men-made problems, we can provide the spare parts, but you need pay the money. and our engineer will guide you to operate and use it through the phone or the problems is so complex, we will send the engineer to your local site to help you solve it.