Electromagnetic flow meter Installation suggestion
In order to avoid measurement errors caused by gas entrapment and damage to PTFE and rubber lining caused by vacuum,
It should be installed at the lower part of the horizontal pipe and vertically upwards to avoid installation at the highest point and the vertical point of the pipe.
Should be installed at the rise of the pipe
Install the pipe in the open drain and install it in the lower part of the pipe
If the pipe drop exceeds 5m, install an exhaust valve downstream of the flowmeter.
Control valves and shut-off valves should be installed downstream of the flowmeter and should not be installed upstream of the sensor
The flowmeter must never be installed at the inlet of the pump and should be installed at the outlet of the pump.