The float level gauge should be mounted in a position where the float does not interfere when it moves up and down.
· The measured medium should not contain ferromagnetic impurities.
· In order to prevent the floating ball from falling or colliding on the sealing tube during transportation, causing the measurement belt to come out,
The floating ball is fixed at the bottom of the sealing pipe with a clamp and the magnet is at the top of the sealing pipe. The user should remove the clamp before installation and use, raise the float ball to the top of the seal tube, then gently lower the float ball, and the needle should be rotated accordingly.
• The float level gauge's surface pull indicator is an index type. When the fullness is measured, the indicated number is the direction of the gel needle. When the air level is measured, the indicated number is arranged in the counterclockwise direction.
• Float level gauges are shipped with mounting flanges. Users should have pipe flanges.