Radar level gauge is a common measurement instrument. The measurement principle of radar level gauge is better, and generally no problem occurs during the measurement process. Nowadays, the range of radar level gauges is very wide. Everyone must pay more attention to the installation of radar level gauges. Only when they are installed, the future use will be more secure.
The radar level gauge should pay much attention when it is used for measurement. Its measurement range is limited, and it usually starts when it touches the beam. However, under special circumstances, if the tank bottom of the radar level gauge is concave, it must start from its lowest point. If it is lower than this point, there is no way to measure it.
When using a radar level gauge, pay attention to how much the dielectric constant of the medium is. If it is lower than the liquid level, the bottom of the tank can be seen, and the accuracy of the measurement can be guaranteed. Generally, we recommend setting the zero point of the radar level gauge at the position of C, so that the measurement is more accurate. In theory, radar level gauges have no way to measure the position of the tip. It can be affected by external factors.
These are the items that radar level gauge should pay attention to when it is used in measurement. To make its maximum measurement performance come out, it is necessary to pay more attention when measuring, and it is necessary to grasp the correct measurement method.